Internal fixation of midshaft clavicle fracture using Synthes superior clavicle plate
Subscribe to get full access to this operation and the extensive Shoulder & Elbow Surgery Atlas.
Learn the Internal fixation of midshaft clavicle fracture using Synthes superior clavicle plate surgical technique with step by step instructions on OrthOracle. Our e-learning platform contains high resolution images and a certified CME of the Internal fixation of midshaft clavicle fracture using Synthes superior clavicle plate surgical procedure.
Clavicle fractures are common accounting for around 2.6% of all fractures, with middle third fractures accounting for 80%. 2-6% occur in the medial third, while the remaining 12-16% occur in the lateral end of the clavicle. These fractures most commonly occur in a young and active population.
The management of such injuries can be controversial as historically, displaced mid-shaft clavicle fractures have been managed conservatively with low reported non-union rates. However, more literature has reported much higher rates, up to 20% on occasion. It is widely accepted that this can be reduced significantly with surgical intervention.
Whilst there is some risk with any surgical intervention, the published results also demonstrate that plate and screw fixation can be performed safely to give a good outcome with improved patient satisfaction and a reduction in the non-union rate compared to conservatively managed fractures.
For displaced lateral end clavicle fractures, there is a higher rate of non-union and therefore, the threshold for surgical fixation is lower. Options for surgical fixation include specific lateral end anatomic clavicle plates utilising a variety of locking screw options for the smaller lateral fragment. Some plates allow reconstruction of the ruptured coracoclavicular ligaments with a non-absorbable material to augment and support the repair. If standard internal fixation is not possible due to comminution, these fractures can also be managed with a hook plate. The lateral end of this plate hooks under the acromion and butresses the lateral clavicle from displacing superiorly. The disadvantage with this type of fixation is that it needs to be removed once the fracture has healed to avoid impinging on the rotator cuff and causing a cuff tear. Whilst in situ, it may also physically limit the arc of movement in the shoulder.
There are many implant companies with clavicle specific plates. My preference is for the Synthes LCP clavicular (Locking Compression Plates). These are anatomically contoured and sided plates with either a superior or supero-anterior plates option for mid-shaft clavicle fractures. Superior clavicle plates tend to require less contouring. Theoretically, the medial screws are also directed further away from the neuro-vascular bundle. The plates and screws on the Synthes set are made of stainless steel with the “combi-hole” design allowing the option of either locking or non-locking screws within the same hole. The Synthes clavicle set also provides useful instruments to facilitate the surgical exposure and fracture reduction.
Author:Mr Sam Chan, FRCS (Tr & Orth).
Institution: The Queen Elizabeth hospital, Birmingham, UK.
- Each operation and the questions associated become a named course in the CPD section
- The operative technique itself is read as a lesson as is any company implant information if this is being assessed.
- You’ll need to tick the box to confirm this has been done and can do this immediately if you have already read the op tech.
- The vast majority of operations have a 10-15 MCQ quiz covering all aspects of the decision making and the technique
- There are four possible answers of which one is correct (or on occasion more correct) than the others.
- There are additional quiz modules on the surgical steps, the implants and problem cases being added continually
- The course is completed once all the lessons are read and quizzes submitted and passed.
- On successful completion of each quiz you will receive validated CPD points that add to the certificate in your CPD folder.
- Your dashboard also will contain a record of the time you have spent logged onto and using the site.
- The timer suspends after 5 minutes though if there is no activity.
- When you restart you will resume at the same point in the module.
- Once you have completed each quiz you will need to feedback on the module first then click “submit” and your paper will be marked.
The pass mark is 75%.
- If you fall below this level you will be directed back to re-read the slides where you’ve tripped up.
- Once these have been read you can re-do just the questions you failed on.
- Once you have passed the quiz you can return at a future stage & resit .
- Operation Quiz – 1 CPD point
- Surgical steps Quiz – 1/4 CPD point
- Implants Quiz – 1/4 CPD point
- Problem case Quiz – 1/2 CPD point
One CPD point equates to one hour of academic activity
Welcome to the Professional Development question section. The objective of taking these tests is to demonstrate that you have understood all aspects of the assessment and management of patients requiring surgical intevention. On successful completion you will receive a certificate accredited by both the Royal College of Surgeons of both England and Edinburgh as well as the British Orthopaedic Association.
Our content is designed for both Surgeons in independent practice and Surgeons in training.