Intertrochanteric neck of femur fracture treated with a proximal femoral nail (Synthes long TFNA)
Subscribe to get full access to this operation and the extensive Hip Surgery Atlas.
Learn the Intertrochanteric neck of femur fracture treated with a proximal femoral nail (Synthes long TFNA) surgical technique with step by step instructions on OrthOracle. Our e-learning platform contains high resolution images and a certified CME of the Intertrochanteric neck of femur fracture treated with a proximal femoral nail (Synthes long TFNA) surgical procedure.
Intertrochanteric hip fractures are more commonly treated with a Dynamic Hip Screw (DHS) in the UK. However, there is good reason to consider a nail in certain situations, in particular if there is any lateral wall or greater trochanter compromise. This is because the DHS is designed to allow the fracture to collapse and in the absence of a lateral wall to buttress this collapse, the femoral diaphysis can medialise and the hip abductors lose their mechanical advantage. If recognised, this situation can be prevented by augmenting the DHS with a Trochanteric Stabilisation Plate (TSP). However this often irritates the lateral soft tissues and can cause a trochanteric bursitis.
The National Institute for Health and Care Excellence (NICE), an advisory body in the UK, states that adults with trochanteric fractures above and including the lesser trochanter should be treated with extramedullary implants. Evidence to support their position is that similar clinical outcomes can be achieved with both types of device and there is some evidence for a higher re-operation rate due to periprosthetic fracture with a nail, which is also a more expensive implant than a DHS. Looking outside of the UK, then these fracture patterns are routinely treated with intramedullary devices.
Intramedullary implants have been reported to be associated with a higher 30 day mortality when compared with sliding hip screws and Whitehouse et al. suggest that for every 112 femoral nails there will be 1 excess death. However, they do not know whether the implant / technique is causative or whether there were significant differences between the cohorts, as this was not a randomised controlled trial. It is therefore important to ensure that these devices are used selectively.
The TFNA is available as either a short nail (lengths 170mm, 200mm, 235mm) or a long nail (lengths 300-480mm) and can be used with either a screw or helical blade. The proximal end of the nail is 15.66mm and distally the nails are available in 9mm, 10mm, 11mm, 12mm and 14mm diameters. It has a radius of curvature (ROC) of 1.0 metres which is different to its’ predecessor the PFNA which had a ROC of 1.5 metres. Internally there is a pre-assembled locking mechanism to control the blade rotation and either lock it statically or dynamically. The blade angles are available in 125 degrees, 130 degrees and 135 degrees and all nails have an anatomic 10 degrees of anteversion. The nails are made from titanium and accept 5mm locking bolts distally. In comparison with a competitor nail such as the Stryker Gamma3 nail which has a ROC of 1.5 metres, the TFNA is more likely to have an anatomical fit with the normal anterior bow of the femur and reduces the risk of anterior impingement of the nail in the distal femur which can potentially perforate or cause distal femoral fractures.
Implant failure is a rare event but Synthes has recognised that when these nails do fail, they commonly tend to fail at the screw / blade hole. This can mean that subsequent extraction is difficult. To overcome this there is a thread internally in the nail (below the blade hole) where an instrument can be attached and the nail removed more easily.
The TFNA can be augmented with low viscosity bone cement that can be injected through the cannulated screw or helical blade into the femoral head for patients with severe osteoporosis. To do this, the insertion handle is removed and the 3.2mm guide wire is removed from the screw or blade. The cement is then injected through the cannulation where the wire previously was and exits through perforations in the tip of the screw / blade. Bone cement (polymethylmethacrylate – PMMA) is however contraindicated in the following circumstances:
- If there is a risk it will leak into the joint e.g. perforated the head with a guide wire
- Intracapsular fractures
- Patients with severe cardiac / pulmonary insufficiency
- Patients with known allergy to any of the cement components e.g. antibiotics
Whitehouse MR, Berstock JR, Kelly MB, Gregson CL, Judge A, Sayers A, Chesser TJ. Higher 30-day mortality associated with the use of intramedullary nails compared with sliding hip screws for the treatment of trochanteric hip fractures. A prospective national registry study. Bone Joint J. 2019 Jan; 101-B(1): 83-91.
Readers will also find the following OrthOracle operative techniques of interest:
Author:Mr Ross Fawdington FRCS (Tr & Orth)
Institution: The Queen Elizabeth Hospital, Birmingham, UK.
- Each operation and the questions associated become a named course in the CPD section
- The operative technique itself is read as a lesson as is any company implant information if this is being assessed.
- You’ll need to tick the box to confirm this has been done and can do this immediately if you have already read the op tech.
- The vast majority of operations have a 10-15 MCQ quiz covering all aspects of the decision making and the technique
- There are four possible answers of which one is correct (or on occasion more correct) than the others.
- There are additional quiz modules on the surgical steps, the implants and problem cases being added continually
- The course is completed once all the lessons are read and quizzes submitted and passed.
- On successful completion of each quiz you will receive validated CPD points that add to the certificate in your CPD folder.
- Your dashboard also will contain a record of the time you have spent logged onto and using the site.
- The timer suspends after 5 minutes though if there is no activity.
- When you restart you will resume at the same point in the module.
- Once you have completed each quiz you will need to feedback on the module first then click “submit” and your paper will be marked.
The pass mark is 75%.
- If you fall below this level you will be directed back to re-read the slides where you’ve tripped up.
- Once these have been read you can re-do just the questions you failed on.
- Once you have passed the quiz you can return at a future stage & resit .
- Operation Quiz – 1 CPD point
- Surgical steps Quiz – 1/4 CPD point
- Implants Quiz – 1/4 CPD point
- Problem case Quiz – 1/2 CPD point
One CPD point equates to one hour of academic activity
Welcome to the Professional Development question section. The objective of taking these tests is to demonstrate that you have understood all aspects of the assessment and management of patients requiring surgical intevention. On successful completion you will receive a certificate accredited by both the Royal College of Surgeons of both England and Edinburgh as well as the British Orthopaedic Association.
Our content is designed for both Surgeons in independent practice and Surgeons in training.