Ankle replacement-Wright Prophecy surgical technique
Subscribe to get full access to this operation and the extensive Foot Surgery Atlas.
Learn the Ankle replacement-Wright Prophecy surgical technique with step by step instructions on OrthOracle. Our e-learning platform contains high resolution images and a certified CME of the Ankle replacement-Wright Prophecy surgical procedure.
Customised Patient Specific Instrumented ankle replacements are the latest advance in this emerging field.
The first ankle replacement was implanted in 1970 by Lord & Marotte. It was effectively an upside down hip stem and naturally the results were poor. Since that time there have been several generations of implants. Initially the implants were constrained and cemented, later followed by unconstrained using a mobile bearing meniscal implant and the most modern of implants are now two component fixed bearing.
Ankle replacement technology is an emerging but rapidly growing field. Approximately 900 ankle replacements are performed in the UK each year, in comparison to perhaps 2000 ankle fusions (NJR and HES Data).
The basic premise for ankle replacement technology is based on a lack of understanding how the joint actually works. Initial concepts that the joint is a monoaxial hinge are of course incorrect. Aside from just plantarflexion and dorsiflexion, the joint rotates, inverts and everts and translates forwards and backwards. Indeed even sagital plane motion is not just in one plane as the arc of motion of the talus on the lateral side is greater than the medial side hence creating a rotatory motion within the mortise by the frustrum shaped talus.
Modern thinking and newer generations of implants have removed the need for cement, using porous coatings or HA coatings for osseointegration, ultra high moleccular weight polyethylene (UHMWPE) as an articulating surface, and better instrumentation. Most implants use cobalt chromium allow although some implants are made also of titanium. The most recent innovation was the use of patient specific instrumentation (PSI) which enables CT scan data of the patient to produce patient specific jigs from nylon that can be used to plan the surgery in advance avoiding the need to determine variables such as alignment, size and rotation on table.
The INFINITY™ Total Ankle System is produced by an American company Wright Medical. Several surgeons were brought on to provide a clinical perspective and the resulting replacement has benefitted from the collaboration between themselves and Wrights’ design engineers.
The Infinity ankle consists of a highly polished metal talar dome, a titanium alloy tibial tray, and an UHMWPE bearing surface.
The key features they claim to have introduced are:
- Low-Profile tibial implant design
- Resurfacing talar component
- Fluoroscopic Visualization
- Talar Component Interchangeability with other revision implants
- Optimized to use PSI preoperative navigation.
The tibial tray design is low profile and has 3 pegs which are angled backwards so that it can be inserted and impacted from the front. It requires the same standard resection depth for all implant sizes. It comes with a standard and long option to allow for optimal AP coverage.
The spacer or meniscus is made of UHMWPE which is fixed into the tibial component. This seems counter intuitive to the European mobile bearing implants where the meniscus is not fixed but instead is mobile between the joint surfaces. However, there are several reasons they have gone for the fixed bearing solution. Firstly, the FDA (the US regulatory body) does not recognise mobile bearing implants but does recognise fixed bearing implants. Second, the evidence to demonstrate a difference in performance between mobile and fixed bearing implants in knees isn’t there and indeed a small non randomised study by surgeons involved in the Salto Implant (Tournier) failed to show any advantages of mobile bearing over fixed bearing menisci.
Obviously, many of the elements of the Infinity’s design had been used before by other implants on the market and the real innovation in my mind was the ability to use PROPHECY PSI to plan and perform the surgery.
Author: Andy Goldberg MD FRCS (Tr & Orth)
Institution: The Royal National Orthopaedic Hospital, Stanmore, UK
- Each operation and the questions associated become named course in the CPD section
- The operative technique itself is read as a lesson as is any company implant information if this is being assessed.
- You’ll need to tick the box to confirm this has been done and can do this immediately if you have already read the op tech.
- The vast majority of operations have a 10-15 MCQ quiz covering all aspects of the decision making and the technique
- There are four possible answers of which one is correct (or on occasion more correct) than the others.
- There are additional quiz modules on the surgical steps, the implants and problem cases being added continually
- The course is completed once all the lessons are read and quizzes submitted and passed.
- On successful completion of each quiz you will receive a validated certificate in your CPD folder.
- Your dashboard also will contain a record of the time you have spent logged onto and using the site.
- The timer suspends after 5 minutes though if there is no activity.
- When you restart you will resume at the same point in the module.
- Once you have completed each quiz you will need to feedback on the module first then click “submit” and your paper will be marked.
The pass mark is 75%.
- If you fall below this level you will be directed back to re-read the slides where you’ve tripped up.
- Once these have been read you can re-do just the questions you failed on.
- Once you have passed the quiz you can return at a future stage & resit .
Welcome to the Professional Development question section. The objective of taking these tests is to demonstrate that you have understood all aspects of the assessment and management of patients requiring surgical intevention. On successful completion you will receive a certificate accredited by both the Royal College of Surgeons of both England and Edinburgh as well as the British Orthopaedic Association.
Our content is designed for both Surgeons in independent practice and Surgeons in training.